Defibrillator cable & AED Electrodes Pads Cable
Defibrillator Coil & Defibrillator wire
Defibrillator wires have 3 main forms：Coil & Round;Straight & Round and Flat Ribbon.The most used is Coil cord for AED paddle,and Flat Ribbon cable for AED Pad.
The Defibrillator cable usually contains 3 to 10 depends on its purpose and applications. A simple 3 wire defibrillator (it could be a aed training machine) only need 3 wires for the paddle. And ECG Defibrillator will use 4-8 signal wire to get ECG signal,indicates the operator if it is necessary to make difibrillation and when to make it. The more Intelligent a ECG Monitor with defibrillator is,the more wires it need.ECG Defibrillator monitor the heart and indicates the operator when is the best time to defibrilate and how many joules power is needed.It make defibrillation easier and more accurate.
As a China Medical cable manufacturer ,we custom AED cable ( AED leads) to interconnect between its power unit and the patient paddle or Pads according to customer,s requirement
AED electrode pads become a mainstream，compared with Paddles,AED Pads/Gel are used to reduce transthoracic impedance when paddles are applied directly to the chest wall to deliver a shock. Better protective for patients and doctors. Especially with Mindray intellectual AED, an un-trained person can also make an emergency rescue under the guide of the defibrillators’ screen and voice
Trunk Defibrillator Cable and Branch Defibrillator cable for AED Paddles
Considering the convenience of collection and extension of the Defibrillator cable for a AED paddle,it is usually in the form of Coil cord,which is retractible. To learn some knowlege about how long can a coild cord can be stretched is very important ,as there will be Voltage Drop with the length of the cable. An unsufficient voltage may lead to a defibrillation failure.
A typical Trunk defibrillator cable consists of 2 High voltage wires (usually need to withstand 3000-5000 V DC voltage) and several signal wires with or without EMC shielding. A typical Branch defibrillator cable usually with one high voltage wire and same number signal wire with the trunk cable
Traditional Defibrillator cables usually does not pay attention to the signal wire,there is no any shielding between high voltage wire and signal wires.
AED leads for Electrodes Pads & Gel
In public places,such as metro station,airport,With the development and wide application of ECG Defibrillator ,Defibrillator manufacturers pay more attention to the ECG signal.such as in the latest zoll onestep cable,there are 3 separate ECG lead cable for defib with grahene coating (which is to assure a Low Noise performance,see the right picture :Flat ribbon cable)
difibrillator voltage and Difficulty In AED Cable design & production
As a main first aid life saving devices, Zero-failure is the common will for the paddle/Pads manufacturer and defibrillator cable wire manufacturer. Any Lead failure may lead to death or heart damage.PS(:Lead failure was defined as: failure to capture/sense, abnormal pacing and/or defibrillator impedance, visual insulation defect or lead fracture, extracardiac stimulation, cardiac perforation, tricuspid valve entrapment, lead tip fracture and/or lead dislodgment)
The defibrillator voltage is usually know as 3000-volt charge in less than 0.001.Some brand clain 200-1000 voltage and somee brands clain 5000-volt charge in less than 0.001 .If you are not sure on the voltage,you may focus on the Joules (Monophasic Waveform Defibrillators 360J,Biphasic Waveform Defibrillators 150J-200J ) .Voltage drop is another key factor for AED cable. a common AED paddle cable has 3-5 meters stretched length(it will be 40-50cm after coiling,for easy collection). In Some ICU,a AED cable stretched length may even reach 9 meters(in this way,operator do not need to move the defibrillator ,and can fast defibrillate because of the long range of the AED electrode). Defib Energe insufficience will lead to difibrillation failure,further damage to heart ,even death
What we do to conquer these challenge faced with defibrillator pad
As a cable partner for two of the world top 4 AED brand, the cable we developed for AED electrodes and defib pads are always exceeded their requirements.
Considering some special situation, such as a Patient who may need rescue at any time, so a Defib pads may with him/her 24 hours. During this period, he/she may need an imaging, like X-ray, metal should be removed.so we design a Carbon fiber conductor based AED leads. Carbon fiber is almost invisible on x-ray and other imaging devices.
Non-metals, such as Carbon Fiber (Carbon fiber is radio translucent, making it an excellent material for EKG leads and defibrillators.)
Defibrillator pad cable Features
Conductors Materials for Defibrillator cable Leads
- Carbon Fiber
- Tinned Copper
- Silvered copper
- Alloy Copper
Insulation Materials for Defibrillator cable Leads
Shielding Methods & Materials
- Conductive tape
- Aluminum-Mylar Foil
- Tinned/silvered Copper Braiding
- Graphene Mixture ink coating
Defibrillator pad cable Manufacturing procedure
you maybe also need following customized AED cable
Teflon Coiled cable AE205x Medical class coiled cable for” Automated External Defibrillator” Paddle Features Telfon Insulator for all the wires,bright TPU Jacket 1*20AWG high voltage wire 4*24AWG singla wire technical specification 1:Specification：134/0.07BC*1C+60/0.07BC*4C+PTFE 5.2 MM TPU
this 1+4 core coil wire is the trunk cable developed to interconnect the Power unit and Paddle of the AED.1 High voltage wire coated with Aluminum foil lined with drain wire ,and 4 core for signal.
this 2+7 core coil wire is the trunk cable developed to interconnect the Power unit and Paddle of the AED.2 High voltage wire coated with Aluminum foil lined with drain wire ,and 7 core for signal.
This Flat AED cable used to interconnect the Power unit and Electrodes Pads/Gel of ZOLL defibrillator.AED Pads is more and more becoming a mainstream in emergency rescure heartstart